Mobile c-arm x-ray machine is commonly used in orthopedic surgery, the equipment for the operator’s technical requirements, even professional physicians need to be trained in order to operate skillfully. Today we share with you a common problem in the use of c-arm machine – how to adjust the window size of mobile c-arm x-ray machine.
I. Turn on the machine
(1) Connect the whole c-arm machine to the Internet, close the machine air switch, click the mask power on button, click the UPS power switch, and then click the computer host power on button.
(2) Double-click the computer desktop icon [CArmWor…. …], start the workstation software, enter the workstation software according to the patient registration interface form prompts, fill in the patient information, and then click “Register and Save” to enter the image acquisition interface.
Click the automatic fluoroscopy icon, and then click the required irradiation site, such as irradiation of the patient’s lumbar spine, click the lumbar spine part of the human body graphic, and then step on the foot brake 3S – 5S, can be imaged;.
Click the manual fluoroscopy icon, then click the required irradiation site, such as irradiation of the patient’s lumbar spine, then click the human body graphic lumbar vertebrae parts, and adjusted to the appropriate “KV mA” value, and then step on the foot brake 3S-5S, can be imaged.
Click on the pulse fluoroscopy icon, then click on the desired irradiation site, such as irradiation of the patient’s lumbar spine, click on the human body graphic lumbar vertebrae, and adjust to the appropriate “KV mA” value, and then step on the foot brake 3S-5S, can be imaged.
If you want to adjust the size of the window of the mobile c-arm x-ray machine, you need to find the “magnifying glass” button in the toolbar, as shown in the figure below, click on the image window can be zoomed in and out, have you learned?